- Autorefracts patients of all ages without dilation (6 months to seniors)
- Spherical equivalent range measures +/- 7.50D
- Auto-capture technology provides instant results
- Connected workflow can sends results directly to your EMR •
- Measures pupils down to 3 mm (20 to 100-year-old)
- Refracts monocular or binocular (SE, DS, DC, axis)
- Touchscreen display is easy to use
- Results for both eyes are displayed on screen in seconds
Easy to Use
Bring the Exam to Your Patients
When treating patients with small pupils, limited mobility and disabilities, it can be hard for eye care professionals to maintain efficient workflows. The handheld, portable Spot Vision Screener can make identifying refractive errors easier and more efficient.
Education & Documentation
Get in the know to get the most value out of your solution.
How to Videoskeyboard_arrow_down
Screen a Subject
Connect to a Wireless Network
Connect to a Printer
Export Screening Results to a Computer
User Manual (Previous Versions)keyboard_arrow_down
Frequently Asked QuestionsExpand all
How long does it take to fully charge the Spot Vision Screener battery and how long does the charge last?keyboard_arrow_downOnce the battery is fully depleted, recharging takes two hours. The battery will last over four hours on a single charge.
What is included with my Spot Vision Screener?keyboard_arrow_down
You will get a carrying case, cleaning cloth, instructions for use and an accessory box that includes:
- Quick Start Guide
- AC power cord
- Power supply/charger
- Hook-and-loop cable straps
Damages incurred by shipping the device in non-approved packing material will void the limited warranty. Do NOT store your vision screener in the case while the power cord is plugged in.
Does my Spot Vision Screener require calibration?keyboard_arrow_downNo, your screener will arrive calibrated as part of the manufacturing process and won’t require calibration from that point.
When the Spot Vision Screener is fully charged, how many screenings can occur on average before needing to be charged again?keyboard_arrow_downThere is no set number of screenings per charge due to variation in screening time needed for each particular patient; however, your screener should operate for over four hours before requiring a recharge.
How do I update the software?keyboard_arrow_downThe unit must be returned to our service facility for a software update. It is free of charge (including shipping) by contacting customer support.
What is the warranty on my Spot Vision Screener?keyboard_arrow_downOne year. Use our Warranty Lookup Tool to check the warranty on your unit.
Where is the serial number located?keyboard_arrow_downIt is located on a label on the bottom of the unit.